Quantus™ Fluorometer Firmware
Download the latest Quantus™ Fluorometer Firmware (version 2.25)
- Download the Quantus™ Instrument Firmware and Firmware Loader by clicking the link below.
- Extract the contents of the download file.
- Press the right and left key pad buttons simultaneously on the Quantus™ Fluorometer, and while pressing these buttons connect the Quantus™ Fluorometer to a USB port on the PC.
- From the extracted download file, open the Firmware Loader by double-clicking the “AT91_CDC-USB_FWLoader” file.
- Click the “Open FW File” button on the Firmware Loader.
- Navigate to the extracted firmware file. This will be called “Quantus – v2.25.bin” or something similar, based on the version.
- Click “Open”.
- Click the “Update FW” and wait until firmware updates successfully.
- Power Off / On the Quantus™ Fluorometer.
- Navigate to the Tools screen. The new firmware number should be displayed.