Slicprep™ Centrifuge Compatibility
The Slicprep™ 96 Device is a single device for high-throughput incubation, centrifugation and separation of DNA samples on a solid support. The device consists of 3 components: a 2.2ml 96 Deep Well Plate, a 96 Spin Basket and a U-Shaped Collar.
The following centrifuge and rotor combinations have been confirmed as being compatible with the Slicprep™ 96 Device (Catalog numbers included where known):
|GH-3.8 swinging-bucket rotor (Cat.# 360581) with MicroPlus
Carrier Assembly (Cat.# 362394)1
|JS-4.3 rotor (Cat.#
362734) with MicroPlus Carrier Assembly (Cat.# 362394)1
Avanti-J26XP, Avanti JE
and Avanti J20
|JS-5.3 rotor (Cat.#
|Eppendorf 5804, 5804 R,
5810 and 5810 R
(Cat.# 5804 740.009)1
|Beckman Coulter Allegra X-12
||SX4750 rotor and Microplate Carriers
for SX4750 Rotor (Cat.# 393042)1
|JS-4.3 swinging-bucket rotor
(Cat.# 362734) with MicroPlus Carrier Assembly (Cat.# 362394)1,3
||H-3S Horizontal rotor (Cat.# 096-029001)2,3
1Verified by Promega scientists.
2Verified by nonPromega scientists.
3These centrifuges are no longer available from the manufacturer.
Known incompatible centrifuge and rotor combinations
|Thermo IEC Centra CL3 centrifuge
||Rotor 244 and 96-well bucket (Cat. # 50221)4|
4Space between the Slicprep™ 96 Device and the inner sides of the rotor is too small. Uneven centrifugation, plate slippage, or breakage of the Slicprep™ 96 Device may occur. This event is customer-verified.
Promega is committed to providing up-to-date information to our customers. Have you tried another centrifuge and rotor combination with the Slicprep™ 96 Device? Contact Promega via e-mail at: email@example.com, and let us know.
Dimensions of the Slicprep™ 96 Device (with U-Shaped Collar inserted):